Titanium alloy with heat treatment
TiAlV-alloy for the production of metallic restaurations. Applications: Crowns and bridges, frames for metal ceramic veneering, cast partials, primary – and secondary parts for combined restorations, implant supra-constructions. Homogeneous, pore-free structure. Biocompatibility, L 929-Proliferation acc. to EN ISO 10993-5, -12: No deliberation of cell toxic active substances. Corrosion resistance, static immersion test acc. to EN ISO 10271 ( max. 200 μg/cm2 x 7d acc. to EN ISO 22674): Ion release 1,41 μg/cm2 x 7d. Metal-ceramic bond strength (3-Pt.-bending test, min. 25 MPa acc. to EN ISO 9693): 37 MPa (Triceram, Dentaurum).
EN ISO 9693/DIN EN ISO 22674, Type 4
State of starting material
Parts are built in a metal powder bed through a laser (or lasers) melting the metal powder layer-by-layer. Synonyms: Selective Laser Melting, SLM, Direct Metal Laser Sintering, DMLS, Laser Cusing
Data source: Concept Laser
ATTENTION: The presented values are provided by the material manufacturers and are only indicative. Realized material properties may differ from these value for several reasons. Additively does not guarantee the correctness of this information in any way.
Simply copy/paste the following web address and either store it or forward it to someone else.